BESA MRI 3.0 - Released
Get the latest version of BESA MRI today and benefit from the newest features!Browse
MMBT-S Trigger Interface Box
New: In-Built LED to Visualise Trigger Markers and Accessible Runtime Mode Switch!Browse
Bringing you World-Leading Solutions for Research in Neurosciences
Search for Products in one of the main categories below:
Passive vs. Active Electrode - Slew-Rate Problem New
It was brought to our attention that a rumour is buzzing around among EEG researchers that there may be an inherent noise problem with active electrodes. The source of this rumour is a paper published by Sarah Laszlo et al. in 2014 (see A direct comparison of active and passive ampliﬁcation electrodes in the same ampliﬁer system). In this paper, the authors compare the performance of one particular brand (not BioSemi) active electrodes with passive electrodes, both connected to a conventional EEG amplifier (also not BioSemi). Both active and passive electrodes setups were tested with low (2 kOhm) and high (50 kOhm) electrode-to-skin impedances.
The authors find that the total noise (250 Hz bandwidth) for passive electrodes is higher with high electrode impedance, whereas the total noise with the active electrodes is equal (and nearly as low as with passive electrodes with low impedance) for both low and high electrode impedances. However, they also determine that the number of sweeps needed to achieve a reliable ERP result is higher with high electrode impedances for both the active and passive setups. Note though, that the number of required sweeps did not show significant differences between active and passive setups.
The paper starts to derail when the authors attempt to explain why in the tests with high skin-to-electrode impedances, the number of sweeps required for reliable ERP is similar for active and passive setups, instead of a lower number of sweeps for the active case as they would expect on the basis of the better total noise results. So although the active electrodes do not perform any worse than passive electrodes, they basically wonder why the performance is not better than it is.
A logical first step in analyzing the above discrepancy would have been to analyze the noise spectra (for example a relatively small band of extra noise with frequencies around the ERP waveform would explain that the total noise number is hardly influenced whereas the number of required sweeps would certainly be affected). Instead, the authors choose to introduce a hypothesis without any support by measurements or references. The authors suggest that the amplifier in the used active electrodes has a slew rate that is limited to such an extent that it causes distortion of the ERP wave, leading to a higher required number of sweeps than would be expected on the basis of the total noise figures.
It is certainly true that slew rate limitations can lead to signal distortion. The effect usually plays a role only with large (several volts) signals at high (several MHz) frequencies. Suggesting slew rate limitations as a problem for EEG signals (amplitudes in the hundreds of uV range and frequencies below a few hundred Hz) sounds far fetched.
How extreme the hypothesis actually is, can be illustrated by some numbers. The paper shows ERP results with a bandwidth of 250 Hz and an amplitude of 6 uV. The required slew rate to acquire this waveform without distortion is approx. 0.01 V/s (min SR = 2*pi*f*Vpk-pk, see for example the Wikipedia page on Slew Rate). Now consider the standard BioSemi ActiveTwo EEG system with active electrodes. At 16 kHz sample rate, this system can acquire a 3 kHz, 0.5 Vpk-pk wave without significant distortion. The complete system (active electrodes, post-amplifier, analog-to-digital converter) therefore has a slew rate of at least 10,000 V/s (or 0.01 V/us in the more usual notation). The standard BioSemi active electrodes are also applied in a high-frequency version of the ActiveTwo system with 262 kHz sample rate. The setup accurately acquires a maximal sine wave of 80 kHz, 0.5 Vpk-pk, leading to a slew rate estimate for the standard BioSemi active electrodes of at least 0.2 V/us. This is a factor of 20 million faster than in the Laszlo hypothesis. Refer to the following result for a 20 kHz, 40 mVpk-pk input signal acquired with standard BioSemi electrodes and a 262 kHz sample rate version of the ActiveTwo system (the dots indicate sample points at 3.8 us intervals). Note the fast rise and fall times of the square wave edges indicating good slew rate performance (the plot shows rise/fall times of approx 10 us, this is a limitation for step responses of the decimation filter in the ADC).
Designers of EEG systems, with or without active electrodes, all make use of a relatively small pool of quite similar Op-amps (building blocks for amplifier circuits) and ADCs provided by a handful of semiconductor manufacturers. These components all offer slew rates far beyond what is minimally required for EEG signals. I am therefore quite sure that our competitors can present results similar as found for the BioSemi setup. This is supported by the input range and bandwidth figures published by various manufacturers. While I would not hesitate to spread doubts about performance aspects of products by BioSemi's competitors, the limited slew rate argument is so unlikely for any currently available EEG system (with or without active electrodes) that I would never consider using it.
In other words: it is doubtful that the particular (non BioSemi) active electrode evaluated in the paper has a slew rate with a factor of 20 million slower than the BioSemi active electrode. Anyway, such an extremely slow slew rate would have been immediately apparent in the waveforms generated by relatively large and fast artifacts in the EEG signal, such as eyeblinks. In addition, the phenomenon would have been straight forward to measure by connecting the active electrodes to a signal generator. The paper does not indicate that any attempt was made to verify the hypothesis with measurements. Given that the particular active electrodes as tested in the paper are quite widespread under EEG researchers, it is hard to imagine that a slew rate limitation of 0.01 V/s would have remained unnoticed for years among these researchers.
The authors make things worse by stating that slew rate limitations are more severe for amplifiers "at very low output impedance". They even refer to BioSemi to support this claim. In reality, BioSemi never has made such a statement and with good reason: it simply is not true. There are numerous amplifier designs with high slew rate and low output impedance, the BioSemi active electrode as presented above is just an example.
The last sentence of the paragraph is the most bizarre. While it is, of course, true that passive electrodes do not have the slew rate limitations associated with amplifier circuitry, the authors overlook that the input stage of an EEG amplifier for passive electrodes also has high input impedance, low output impedance and slew rate limitations. In terms of amplifier noise, input and output impedances and slew rate, the only principle difference between the active and the passive setup is the location of the first amplifier stage. The statement that "slew rate [..] is simply not an issue for passive electrodes" is therefore surprising. In reality, both active electrodes and input stages for passive electrodes have the same slew rate limitations, although as outlined above, it is unlikely that any EEG system (active or passive) using currently available electronic components will present a problem in this respect. Instead, the authors suggest that slew-rate limitation is an inherent problem encountered with all active electrodes (regardless of brand and design), whereas the effect is totally eliminated in all amplifiers for passive electrodes.
To summarize: the paper presents an unlikely hypothesis that is not verified and not supported by any valid reference. The authors fail to do the simple measurements or numerical evaluation that would have readily falsified the hypothesis. Nevertheless, they find themselves qualified to apply this questionable hypothesis not only to the particular active electrode evaluated in the paper but to the active electrode design in general. In effect, they disqualify all active electrodes, including the successful BioSemi design, and confuse the scientific discussion about the merits of the active electrode principle. It is peculiar to see that such an unjustified accusation, that may affect legitimate manufacturers who put a lot of effort in improving EEG acquisition systems, can make it through the review process of Journal of Neuroscience Methods.
This is an article published by Coen Metting van Rijn, PhD, director, BioSemi, 5-December 2019
EL-CHECK: A vital addition to any MP160 order! New
EL-CHECK measures the impedance between surface electrodes attached to a subject so you can detect and correct any electrode contact issues before you start recording data.
Works on all signal types—measure electrode impedance from <5kΩ to >50 kΩ for all surface biopotential measurements, including ECG, EEG, EGG, EMG, EOG, Bioimpedance, and Impedance Cardiography.
EL-CHECK permits simultaneous connection of up to three electrode leads, for quick impedance checking between any two electrodes in the three connected leads: Active (Vin+), Reference (Vin-), and Ground (GND). Accepts standard Touchproof and BioNomadix leads.
FAQ: How shall I disinfect EEG caps and electrodes and what kind of disinfection solution shall I use after each recording sessions? New
Soap is a lipophilic substance, meaning that it has the ability to chemically bind to fatty/oily substances. The weakest link of the novel SARS-CoV-2 (a.k.a. COVID-19 virus) is the lipid (fatty) bilayer membrane. Due to this fact, soap will resolve the fatty outer layer of the virus and it becomes inactive. We advise cleaning the head-caps with warm water and soap. A shampoo optimized for delicate fabric is advised, such as Ivory Shampoo (which can be purchased from our online shop: https://shop.neurospec.com/ivory-electrode-cap-shampoo-750ml), or similar. When cleaning, use a soft brush to clean and degrease the inside of the electrode holders.
According to literature, alcohol (ethanol or isopropanol) makes the virus inactive within 30 seconds. The electrodes can be submerged in alcohol without damage. Hence, submerging the electrodes for five minutes in alcohol should be the most effective way to prevent COVID-19 infections. Furthermore, the U.S. Food and Drug Administration (FDA) recommends using alcohol-based disinfectant solutions with greater than 60% ethanol or 70% isopropanol.
BESA Celebrating 25 Years New
With the launch of BESA Research 7.1 just around the corner and BESA GmbH celebrating their 25th anniversary, it's the perfect time to celebrate. We want to congratulate BESAS GmbH on their success story and wish them all the best for the years to come. Being one of the biggest and most beloved software manufacturers globally we want to make sure you have the chance to experience and fully utilise this incredible software.
So for a limited time, we want to offer you an amazing discount when upgrading your BESA Research to the newest version 7.1. If BESA Research isn't your preferred software get in touch with us today to get your quote for an upgrade on other BESA software such as BESA Connectivity Package, BESA Epilepsy, the BESA MRI Package or the BESA MEG Add-On to Research.
BESA MRI 3.0 released New
BESA MRI 3.0 has just been released and we are happy to announce the several new features to enhance your data analysis and results review. These new features include:
- Automated Boundary Element Model (BEM) calculation for use in BESA Research source modelling
- Sufficient automated algorithm for placing electrodes recorded in standard 10-10 positions onto individual head surface reconstructed from MRI
- User-configurable multi-slice view of MRI images
- Several brain atlases available for display on structural MRI in a single view or multi-slice view
- Display BESA Research discrete source analysis solutions in individual MRI with confidence limit of source location
- Support for lossless .jpg compressed DICOM files
- Improvements in co-registration visualization and user interaction
The latest BESA MRI 3.0 runs in combination with BESA Research 7.1 and 7.0. Individual head models (BEM, FEM) generated with BESA MRI 3.0 can be imported from BESA Research 7.x for EEG or MEG source analysis. EEG / MEG data can be co-registered with individual MRI data using individually digitized electrode positions or head shape models.
Get the latest BESA MRI 3.0 license directly from the NEUROSPEC online shop. Visit our online store today to get the latest version of BESA MRI today!
Nerve and Muscle Sonography Symposium - CANCELLED New
Join us for this limited event at Spital Thurgau in Münsterlingen. Together with Digitimer, we will be in part sponsoring this gathering.
The Nerve and Muscle Sonography symposium will take place at the Kantonsspital Münsterlingen from 20-21 March 2020. Together with leading neurologists and physicians from Switzerland and Germany, you will dive into the fascinating world of Nerve and Muscle Sonography. This symposium comprises of lectures on both basic and advanced topics. Further, there will be hands-on training where you will get the chance of testing our stimulators model D185 from Digitimer. Get in touch with us to register today as the maximum number of participants is set to 40 people!
IMPORTANT UPDATE: Due to the current unforeseen spreading of the Corona Virus in Switzerland, the symposium has been cancelled. Contact us if you wish for further updates concerning the symposium.
AS4SAN 2020 UPDATE New
With the unfortunate global situation concerning the Covid-19, we are sad to inform you that the AS4SAN committee has decided to postpone the conference until further notice. We are all in this situation together and hope that you stay safe. More updates will come as soon as we know more.
For more information keep an eye on the homepage of the AS4SAN conference.
Live Demo: BSL Home in Action New
Instructors are exploring all options for the fall and we hear a lot of frustration every day. There are many ways to approach teaching practical labs in this era - including going back to using clay models and pipe cleaners - but why not use a technology that will help you manage the unknown and unpredictable in a modern way? BIOPAC will demonstrate how students can continue learning through hands-on practical physiology labs regardless of social distancing, limited class sizes, or campus closures.
BIOPAC offers a human physiology lab kit, BSL Home, that is very easy for instructors to adopt and, more importantly, for students to use. Lessons on EMG, ECG, EEG, EOG, respiration, biofeedback, and aerobic exercise are pre-configured and included with a handheld device that includes integrated hardware and software. Students simply plug in the device and follow the prompts. Instructors benefit from a curriculum run by thousands of labs. Whether students are spaced out 6′ apart in the lab, social distanced in a separate classroom, at home, or in dorms - instructors can still achieve learning objectives using modern technology.
BIOPAC will host an online presentation to demonstrate BSL Home on July 23rd at 20:00 P.M. CET. Learn how you can provide labs without starting over and without redesigning your curriculum.
- BSL Home Demo: Review current technology for online-hybrid-hyflex labs
- Preserve your core human physiology lab learning objectives
- Strategies for holding practical labs with students in different locations
- Open Q&A
New BIOPAC Respiration Data Webinar New
Knowing how to record great respiration data with the right tool and technique is is crucial for experiments. Being one of the major signals that researchers track in their physiological research this webinar will prepare you to get the most out of your work.
Join Alex Dimov from BIOPAC to learn how to record great respiration data. The main focus of this month's webinar will be the various options of recording respiration data but also the experiment setup and analysis of the collected data.
- Selecting the right technique for application
- Optimizing the devices for recording quality data
- Identifying good and bad data
- Preparing data for analysis and some basic analysis techniques
- Live Demonstration of Lab Equipment!
Thursday, August 27th, 2020 from 17:00 – 18:00 PM CEST.
SOME OF OUR RANGE OF PRODUCTS
BIOSEMI - ActiveTwo
BIOSEMI - bringing EEG and ERP to a new level with the original Active Electrode and ActiveTwo EEG Amplifier.
BESA - Research 7.0
BESA - the leading innovators in digital EEG, MEG and MRI software for research and clinical applications.
SHIMADZU - LABNIRS
SHIMADZU - Next-Generation Optical Brain-Function Imaging with functional near-infrared spectroscopy (fNIRS)